Vertex Velp digital thermoregulator
The instrument can be combined with the most common heating magnetic stirrers and is supplied ready for use, complete with temperature probe and power cable for direct connection to AREX, AREC.X, AREX-6 heating magnetic stirrers.
Thanks to the PW 10 branch element, VTF can be used with all types of heating devices such as water and oil baths, hot plates, thermometers, etc.
The VTF thermoregulator can be used in many applications where precision, reliability and flexibility of use are required, as it is equipped with Fuzzy Logic, which allows the thermoregulation to be automatically adapted to different situations, such as power, load and heat loss.
The highly sensitive probe offers state-of-the-art temperature control performance, with an accuracy of ± 0.5 ° C
Exclusive Fixing System
VTF is equipped with an exclusive and innovative integrated fixing system that facilitates the positioning of the Pt100 probe inside the most common beakers.
The two lateral clamps allow the height adjustment of the VTF, while the friction system ensures its movement on the horizontal plane.
No system will be as comfortable!
Integrated Digital Timer
Another unique feature of the VTF is the integrated timer.
It is in fact possible to select up to a maximum time of 24 hours and 59 minutes, while at the end of the countdown the instrument will automatically switch off the thermoregulation.
The timer mode can be recognized by the alternating flashing of the 2 arrows on the left side of the display.
Fixing on rods: from Ø 10 to 13 mm
Pt 100 stainless steel probe: length 250 mm and Ø 3 mm Display: LCD
Degree of protection CEI EN 60529: IP 54
Power supply: 12 Vdc Weight: 0.3 Kg
Dimensions (LxHxP): 75x145x120 mm
Adjustment range: -10 ... +300°C
Accuracy: ± 0.5°C Resolution: 0.2°C
Timed operation: from 0:00 to 24:59 (h: min)
Remaining time display: possible
Display of maximum temperature reached: possible
Display visualization and automatic switching off of the heating plate in case of: - Probe shorted - Probe interrupted